Fluke A Multimeter DIGITAL | Business & Industrial, Test, Measurement & Inspection, Test Meters & Detectors | eBay!. The listing you’re looking for has ended. Fluke A Multimeter. Condition: For parts or not working. Ended: Jun 08, , AM PDT. Price: US $ Corporation (Fluke) has reproduced this copy of a manual which is no All shipments of JOHN FLUKE MFG. CO. . A A rechargeable Ni Cad battery pack.
|Published (Last):||26 January 2013|
|PDF File Size:||9.74 Mb|
|ePub File Size:||15.63 Mb|
|Price:||Free* [*Free Regsitration Required]|
Unscrew the strain relief on the input cable of the Model 80F-5 and remove the front cover. Depress the AC and mA pushbuttons.
Back to home page Return to top. The oscilloscope should display waveform A. This is proven by the following mathematical calculations. Back to home page. Interest will be charged to your account from the purchase date if the balance is not paid in full within 6 months.
No additional flke charges flluke delivery! Any international shipping and import charges are paid in part to Pitney Bowes Inc. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab.
This item will ship to United Statesbut the seller has not specified shipping options.
Fluke A Multimeter
Alternating Current Adjust the variable resistor on the Model 80F- 15 for a 1. An 01 Option is provided for voltmeters having an 11 megohm input resistance, and an 02 Option is provided for voltmeters having an infinite input resistance 10 3 megohms or greater at null. Please reference the instrument model number when requesting a new shipping container.
Connect the oscilloscope input to TP1. Select the 2 range on the A. Repeat steps 13 through 18 until all parameters are met. Other offers may also be available. Back to home page Return to top.
The oscilloscope display should be similar to waveform B3. Autorange is not working. For parts or not working: The time, called read period, that it takes the capacitor to discharge is directly proportional to the unknown input applied to the A. Input Divider Attention goes in to input divider, which is not very easy in this, as its an autoranging meter, so FET switches are involved in the input divider. And you have access to the internals. Sep 23, Please enter a number less than or equal to 1.
Remove the voltage from the Model 80F-5 high voltage probe.
A fuse, located behind the mA input jack, provides protection against current inputs that exceed 2 amps. People who viewed this item also flkue. This 8004a works on the principle that when a voltage 0840a Source Voltage is applied across series connected resistors the Input Divider resistors and the unknown the voltage drop across each will be proportional to the value of each resistor.
Remove all inputs from the A V-fl input. The Ohms Flukd Voltage is applied to this voltage divider and the voltage drop across the unknown resistance is measured by the A to calculate the value of the resistance.
Contact the seller – opens in a new window or tab and request a shipping method to your location. The A RMS Converter monitors the instantaneous voltage and computes the rms value of the input signal. Physical design of the Model 80F-5 allows direct mating to the input terminals of the Fluke voltmeters. To calibrate the Model 80F, perform the following steps: Each power configuration has a unique model number which must be used when ordering this accessory.
Install the Model 80F-5 on the Fluke voltmeter input terminals.
The second function of the divider is to provide resistors which can be connected in series with the unknown resistance applied to the V-H terminal to form a voltage divider. Here it is, twist and take it out. Select the AC range. Correct timing of these signals is derived from a 1 MHz crystal controlled oscillator.
John Fluke Model 8040A Digital Multimeter
The following information will aid in making more accurate temperature measurements. Learn more – opens in new window or tab.
This item may be a floor model or store return that has been used. I used one of my own little 5V adapter. U14 pin 3, VDC function, 2 range, 2. Three versions of the Model 80F are available. Select the ranges indicated in Table and apply the associated current input to the mA input. Note the A display; it should stop changing as the control reaches full counterclockwise rotation.
A mV and 2V ranges: