Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||11 April 2016|
|PDF File Size:||13.87 Mb|
|ePub File Size:||3.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
Good luck with it Should I be looking at doing it with LabView or something else?
Message 5 of 6. The forums emoticons took over for a bit there. Matt – you’re right, the commands for the A are completely different from the – probably because of its age.
Guess you never know until you try!
Has anyone done anything on interfacing with the A since Matt C’s last post? I have worked with the A mannual, as I recall, the commands are nothing like any other instrument I have seen. Message 2 of 6. Message 1 of 6. I suspect that this difference might be causing my problems.
In particular, after setting up and initiating the instrument, I am initiating the measurement mamual reading it as follows: Would you mind posting the code you used on the A? Most Active Software Boards: I am able to communicate with the device through the GPIB interface, and am also able to retrieve information.
In the code above, MHz is an arbitrary frequency.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at.
I went through the manual before, hp191a when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different. Message 4 of 6. Thanks for your help guys.
Agilent / HP 4191A RF Impedance Analyzer
I think the would probably take the old commands, but the ALSO has new commands. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:. When I gave it a try, though, it worked fine. Sorry, that last line of code there is supposed to read: Message 6 of 6.
IIRC, the sweep parameters were something like this: Message 3 of 6. Accepted by topic author Matt C. If not, what would creating a driver for it entail? You really need the manual. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
It’s 84MB, so I manal you have a fast link. I have not worked with that instrument, but the Agilent web site has the operating and service manual here.